Semiconductor Intelligent Pain Physiotherapy Laser Therapeutic Apparatus
What is the application of our Pain Therapeutic Apparatus?
Intelligent pains therapeutic apparatus has been widely recognized by experts in medical field for its unique performance advantages and clinical effects owing to advanced technology introduced from abroad. This product is mainly used for irradiation therapy to the surface of body to treat various pains, acute or chronic injury of movement system caused by dysfunction, rheumatism, migraine, infectious or non-infectious inflammation. Besides, it can also treat herpes zoster neuralgia and eczema, wound infection, and erysipelas with satisfactory effects.
What is the specification of our Pain Therapeutic Apparatus?
Power in use:AC 220v/50HZ
Working style:Direct irradiation on body surface
Display style:LCD display
Time setting:1-99(min) adjustable
Output medium: GaAlAs
Output wave length:810nm/980nm
Output power(left):810nm/0-500mw adjustable
Output power(right):810nm/0-500mw 980nm/0-200mw adjustable
The instability degree of power:≤±5%
Beam divergence angle:≤40 degrees
What is the detail of our machine?
Click here to get price !!!
How to use our Laser Therapeutic Apparatus?
1. Plug the distant control interlock joint inside the file packet along with the machine into the distant control interlock socket installed at the back of the machine case. Pay attention to the position and direction of the joint when inserting it and screw the dead ring outside the joint tightly when you have plugged the joint.
2. This apparatus uses three-pin 220V power source (in other words, the power source must have the protection of ground wire). Check that there is nothing wrong with the power source before inserting the plug into the power socket.
II. Starting the apparatus
Inset the key into the key jack and rotate 90 degrees clockwise until you see the green indicator lamp (POWER) lights up, which implies the power has been connected well. When you have pressed the button of standby engine, its indicator lamp will light up. In about two seconds, the apparatus will enter standby state where it can start and output. When you press the standby button again, the apparatus will switch into holding state where it cannot output.
Power display:(POWER DISPLAY)xxx(mw).
Time display: (TIME DISPLAY)xx(min).
III. Setting the treatment time
1. Set the treatment time according to different diseases within the range of 1-99 min.
2. Use L/R SELECT key to choose the left road parameters or the right road parameters need to be set. You can adjust them when the figures on screen flash. Having chosen the right or the left road, you can use DIGIT SELECT key to select the treatment time or treatment power need to be set.
3. Having selected the data needed to be adjusted, you can increase the values by INSCREMENT key or decrease it by REDUCTION key, and the value being adjusted will keep decreasing or increasing quickly if you continue pressing the corresponding key.
4. The working time is continuously adjustable within 1~99min and the way of time keeping is countdown. Press the starting key and the device will start working and counting down with 1 minute as time unit. It stops working by itself and sends out a warning sound signal of 30s when the displayed time becomes 00. Press the stop key to stop the warning sound.
IV. Starting and setting the rate of power
1. When you have already set time, you can set the power value, but if the treatment scheme is the same as before, there is no need for you to set it again. During the process of treatment, the set values cannot be changed.
2. Adjusting the rate of power of the treatment apparatus: the power of left road is continuously adjustable within 0-500mw; when the right road’s output wave is 810nm, the corresponding power can be continuously adjusted within 0~500mw, and when it is 980nm, the corresponding power can be continuously adjusted within 0~200mw.
3. Emission output operations
3. 1 Having set the data, just press the START key, and the device will start working by continuously sending out, which is indicated by the sign that the START light goes on and the STOP light goes out. If there is a need of stopping in midway, just press the STOP key.
3. 2 The left road: press START key and STOP key at the same time and the device will send out in impulse mode, which is indicated by the sign that the STOP light goes on and the period START light will go out and go on in turn with a frequency of 10Hz. If there is a need of stopping in midway, just press the STOP key.
The right road: press START key and STOP key at the same time and the device will send out in impulse mode, which is indicated by the sign that the STOP light goes on. The period START light will go on for 4 seconds and then go out for 1 second, and then go on for 4 seconds again. It does this in turn, meaning it sends out for about 4 seconds, and then stop for about 1 second and then sends out for 4 seconds again. If there is a need of stopping in midway, just press the STOP key.
Click here to get price !!!
How to contact us ?
Click to contact us now !!!